Information Notice No. 97-87: Second Retrofit to Industrial Nuclear Company IR100 Radiography Camera, to Correct Inconsistency in 10 CFR Part 34 Compatibility
NUCLEAR REGULATORY COMMISSION
OFFICE OF NUCLEAR MATERIAL SAFETY AND SAFEGUARDS
WASHINGTON, DC 20555
December 12, 1997
|NRC INFORMATION NOTICE 97-87:||SECOND RETROFIT TO INDUSTRIAL NUCLEAR COMPANY IR100 RADIOGRAPHY CAMERA, TO CORRECT INCONSISTENCY IN 10 CFR PART 34 COMPATIBILITY|
All industrial radiography licensees.
The U.S. Nuclear Regulatory Commission (NRC) is issuing this information notice to make licensees aware of a retrofit to the lock of the Industrial Nuclear Company (INC) Model IR100 Radiography Camera, to correct an inconsistency in 10 CFR Part 34 compatibility. It is expected that recipients will review the information for applicability to their licensed activities, and consider actions, as appropriate, to avoid problems. However, suggestions contained in this information notice are not NRC requirements; therefore, no specific action nor written response is required.
Description of Circumstances
The recent retrofit to the safety latch plate (NRC Information Notice 97-35, dated June 18, 1997), which corrected a scenario where the INC Model IR100 radiography camera would not pass the horizontal shock test, as required by 10 CFR 34.20, resulted in the equipment not meeting the requirement that the lock should not be easily removed by unauthorized personnel. INC and the State of California evaluated the problem, and INC took action to correct the situation.
The "Safety Latch Plate" on the IR100 radiography camera has a potentially more "fragile area" (as specified in ANSI N432-1980, Paragraph 8.3.2) than the area previously tested. Apparently, if the Safety Latch Plate was struck directly on the front surface of the plate (with the source in the safe locked position), the plate could be deformed and deemed inoperable. It also appeared that the plate might be susceptible to damage if struck from the top and side. To ensure that this type of potential damage was not possible, a bumper that would prevent deformation was welded to the back of the Safety Latch Plate, and the outer top edge of the plate was angled at a 45-to-50-degree angle, to a depth of approximately 3 mm (0.12 inch). The modified device was tested to ensure compliance with ANSI N432-1980, paragraph 8.2.
In the Safety Latch plate retrofit to the IR100, the Allen screws used to fasten the plate to the device were such that they were easily removed by the user. This is contrary to requirements in 10 CFR 34.20(a) which states that all devices in use meet the requirements of ANSI N432-1980. ANSI N432, Section 188.8.131.52, requires that "[t]he exposure device shall have a lock which is not easily removable by unauthorized personnel." To correct this situation, INC has modified the device design to secure the lock screws by placing a self-adhesive metal foil over the screws. The foil and adhesive are such that the foil is difficult to remove and cannot be removed and replaced without showing signs of damage. The foil will serve to inhibit unauthorized personnel from accessing the screws, and to provide a means to alert the licensee to possible tampering of the lock. INC has indicated that it will contact all owners of potentially affected cameras, on record with it, to notify them of the retrofit, and to retrofit the cameras for them or to provide them with the foil covers and instructions on retrofitting the devices. Attachment 1 contains a copy of the retrofit letter.
All affected cameras should be retrofitted as soon as possible. Licensees should contact INC, or the State of California, to obtain retrofit kits or assistance with retrofit procedures, or to report that they have not received their kits. Affected cameras may be checked for completed retrofits during any inspection.
This information notice requires no specific action nor written response. If you have any questions about the information in this notice, please contact one of the technical contacts listed below or the appropriate regional office.
signed by F.C. Combs for
Donald A. Cool, Director
Michele Burgess, NMSS
|Attachment:||INC Retrofit Letter|
OCTOBER 24, 1997
|SUBJECT:||IR100 EXPOSURE DEVICE LOCK.|
ACCORDING TO TITLE 17 CCR 30332(C)(1) AND ANSI N432-1980,184.108.40.206, EACH RADIOGRAPHIC DEVICE MUST HAVE A LOCK THAT IS NOT EASILY REMOVABLE BY UNAUTHORIZED PERSONNEL. IN OTHER WORDS ONLY KNOWLEDGEABLE OR AUTHORIZED PERSONNEL SHOULD DISMANTLE THE LOCK ASSEMBLY.
TO INSURE THE ABOVE REGULATION, INC WILL INSTALL A HEAVY DUTY FOIL LABEL OVER THE ALLEN SCREW ON THE LOCK BOX (RIGHT SIDE ONLY) THIS LABEL WILL STATE "TO BE REMOVED ONLY BY AUTHORIZED PERSONNEL.
INC WILL INSTALL THE LABELS FREE OF CHARGE ON EVERY DEVICE SHIPPED TO US FOR RELOADING OR MAINTENANCE. IF WE DO NOT RECEIVE YOUR DEVICE WITHIN 120 DAYS FROM THE ABOVE DATE, WE WILL SEND YOU THE LABELS FOR YOU TO INSTALL.
THANK YOU FOR YOUR CONTINUED SUPPORT. IF YOU HAVE ANY QUESTIONS PLEASE CALL INC AT 1-800-711-1982.
VERY TRULY YOURS.
E. W. HUDDLESTON
(NUDOCS Accession Number 9712090210)